Avtomaticheskaya Svarka (Automatic Welding), #12, 2020, pp. 52-57
Modernization of optical microscope and its use to obtain digital images of microstructure of deposited metal
A.A. Babinets, I.O. Riabtsev, I.P. Lentyugov
E.O. Paton Electric Welding Institute of the NAS of Ukraine.
11 Kazymyr Malevych Str., 03150, Kyiv, Ukraine. E-mail: email@example.com
The article analyzes the methods of modernization of optical microscopes to obtain digital images and simplifies their subsequent
analysis while conducting basic metallographic examinations of deposited metal specimens. Two main methods of modernization
were considered: with the help of a camera, equipped with special adapters, which is attached to the tube of the microscope eyepiece
and with the help of a video eyepiece, which is installed instead of a standard microscope eyepiece. The main advantages
and disadvantages of each method were noted. Using MIM-7 metallographic microscope, Canon 650D camera, SIGETA MCMOS
3100 video eyepiece, as well as specimens of microsections with a deposited layer of semi-heat-resistant steel of C – Cr – Mo
– W – V alloying system, comparative metallographic examinations were performed. It is shown that the use of a special video
eyepiece SIGETA MCMOS 3100 allows obtaining better digital images of metal microstructures. As an illustration of the main
advantages of the work, provided by the use of the equipment modernized in this way, the results of metallographic examination
of metal, deposited by electric arc method using flux-cored wire PP-NP-120V3KhMF, were provided. It was experimentally established
that the Toupview software, supplied with the eyepiece SIGETA MCMOS 3100, used in these examinations, allows easy
processing of the obtained digital images, which greatly expands the capabilities of basic metallographic analysis. 10 Ref., 7 Fig.
metallography, optical microscope, video eyepiece, arc surfacing, flux-cored wire, deposited metal, semi-heat-resistant steel
1. Litovchenko, S.V., Malykhina, T.V., Shpagina, L.O. (2011)
Automation of analysis of metallographic structures. Visnyk
KhNU, 960, 215–223 [in Russian].
2. Panteleev, V.G., Egorova, O.V., Klykova, E.I. (2005) Computer
microscopy. Moscow, Tekhnosfera [in Russian].
3. Trankovsky, S.D. (2014) How the microscope operates.
Nauka i Zhizn, 2, 101–104 [in Russian].
4. Hawkins, A., Avon, D. (1980) Photography: The guide to
technique. London, Book Club Associates.
5. Guzhov, V.I., Iltimirov, D.V., Khaidukov, D.S. et al. (2016)
Modification of optical microscope. Avtomatika i Programmnaya
Inzheneriya, 2, 71–76 [in Russian].
6. Lutai, A.M., Klimchuk, O.S., Klyufinskyi, V.B. (2016) Automation
of analysis of metallographic microstructures. In:
Proc. of 3rd Int. Sci.-Pract. Conf. on Automation and Computer-Integrated Technologies. Kyiv, NTUU KPI, 121–123.
7. Glukhova, K.L., Dolgodvorov, A.V. (2014) Examination of
microstructure of composite structural material at the stage
of carbon-filled plastic producing. Vestnik PNIPU. Aerokosmicheskaya
Tekhnika, 2, 222–235 [in Russian].
8. Ternovykh, A.M., Tronza, E.I., Yudin, G.A., Dalskaya, G.Yu.
(2013) ELEMENTIZER – program module of microstructural
analysis. Vestnik MPGUPiI. Priborostroenie i Informatsionnye
Tekhnologii, 44, 106–114 [in Russian].
9. Zubko, Yu.Yu., Frolov, Ya.V., Bobukh, A.S. (2017) Influence
of MECAP on microstructure of AD0. Obrabotka Materialov
Davleniem, 2, 93–100 [in Russian].
10. Lentyugov, I.P., Ryabtsev, I.A. (2015) Structure and properties of metal deposited by flux-cored wire with charge of used metal-abrasive wastes. The Paton Welding J., 5/6, 87-89. https://doi.org/10.15407/tpwj2015.06.19
Advertising in this issue: